福祿克Fluke OptiFiber? Pro?OTDR系列
福祿克Fluke SimpliFiber? Pro光功率計和光纖測試儀
福祿克Fluke Fiber QuickMap?多模光纖距離和故障定位儀
福祿克Fluke Fiber OneShot? PRO光纖測試儀
福祿克Fluke VisiFault? Visual Fault Locator電纜連通性測試儀
福祿克Fluke MultiFiber? Pro光功率計及光纖測試工具包
福祿克Fluke FI-7000 FiberInspector? Pro光纖顯微攝像機
134 1039 1245
If you have access to YouTube, you can click here to watch a video of this new feature in action.
In data centers where the shield is typically grounded at both ends, ANY wire map tester will report the shield as connected, even if it is not. That's because the shield test is a simple DC Resistance measurement. If the shield is open, it will find the remote unit using the building ground (earth).
The DSX CableAnalyzer offers a unique solution to this problem. It ties the four pairs together and then sends a differential signal down the cable treating the shield as the second conductor. This allows the DSX CableAnalyzer to view the shield in the time domain. So even if the patch panels are grounded, it will report the shield open, if indeed it is open you would see:
And since it is using time domain techniques, the distance to the open is given. In this example, we can see the patch panel at the near end is the issue.
The scenario above occurs when the technician grounds the connector on the non conductive side of the cable shield. The result is degraded Alien Crosstalk performance. Since users are told that shielded cabling systems do not need to be tested for Alien Crosstalk, one wonders how wide spread this issue is and whether users are getting the performance they paid for.
If you only use the drain wire to connect the shield and that drain wire is greater than approx. 2 inches (5 cm), the DSX CableAnalyzer may report the shield as open. This is by design. Such a termination acts as an open circuit at the frequencies we are talking about here.
Copyright? 1998-2023深圳市連訊達電子技術開發有限公司 All Rights Reserved .
備案號： 粵ICP備18089363號 網站地圖
免責聲明：福祿克、FLUKE、FLUKE NETWORKS、 NETALLY等品牌資料及圖片摘自網絡，連訊公司不擁有以上品牌商標及相關產品著作權。